Adapter plate VAST, with active ID-Chip, set of 2 (List price: 4.880,00 DKK)
The adapter plate forms the central interface between the probe head and the stylus system. The stylus system is fixed via an integrated cube. Zeiss CFX extensions can be used for optimization. The Zeiss adapter plate is read out and recognized on the probe head when it is picked up. For a CNC-controlled automatic stylus system change, the adapter plate (including the stylus system) is placed in the MT/VAST probe socket (600660-8365-000), which is located on the storage rack (MSR / ProMax).
- ID chip label 100% identification of the correct probe system in the firmware/software to avoid collisions and incorrect measurements.
- Safe handling of the stylus system (no incorrect pick-up possible in the stylus scoket)
- Improvement of system performance (VAST Performance FlyScan QuickChange)
- More information on the control panel display (name, coordinates, measuring force, angle)
- product type
- Adapter Plate
- Connection Type
- Connection Type Out
- 350,0 g
Combinable sensor technology: VAST probe heads (VAST XT Gold and VAST Gold) for probe VAST XTR gold: no rotation functionality (collision protection ring is mandatory) Usable options: VAST Performance Kit FlyScan and QuickChange Combinable probe socket: MT/VAST probe socket 600660-8365-000 A maximum weight of 800 g may be suspended from the adapter plate, (adapter plate included). The length of a stylus system may not exceed 800 mm.
2 VAST adapter plates (600667-9612-000) each one with a cube and an active ID chip Not included in the scope of delivery: Collision protection ring: 600664-8410-000 and labeling clips 626119-0010-000
The features of ZEISS measuring machines and the guaranteed machine specifications are only guaranteed when using original ZEISS exchangeable plates. All functions of the VAST Performance Kit can only be used in combination with ZEISS exchangeable plates. There is a not insignificant negative influence on accuracy and stability if adapter plates from unknown / non-certified manufacturers are used. Check the necessity of performing the DeviceActovator when using a new ZEISS adapter plate. Further information and operating instructions are available at the following link in the Download Center: https://portal.zeiss.com/download-center/softwares/imt